Reference Sodium Analyzers for Steam Drum
Industry Manual Repository
Join the AnalyzeDetectNetwork and Read This Manual and Hundreds of Others Like It! It's Free!
Measurement of Steam Purity. Why use Sodium analyzer after Steam Drum to Super Heater?
Accurate measurement of steam purity is essential to identifying the cause of potential or existing steam
purity problems in modern boiler plants. One reason for this is that superheated steam turbines have an
extremely low tolerance for solids contamination in the steam. Fortunately, techniques are available to
determine steam contamination in the parts per billion range to satisfy the demands of most systems.
The test results make it possible to determine the effect of changing boiler operation on steam purity.
Impurities present in steam can be solid, liquid, or gaseous. Solids are usually dissolved in water
droplets or are present as dust. Because water treatment practices are such that most soluble chemical
constituents of boiler feedwater are converted to sodium salts, most solids present in steam are sodium
salts, with minor amounts of calcium, magnesium, iron, and copper also present.
Gaseous constituents commonly found in low-pressure steam (less than 2000 psig) are ammonia,
carbon dioxide, nitrogen, amines, and silica. Of these, only silica contributes to the difficulties commonly
associated with impure steam; the other constituents are of concern only where they interfere with the
measurement of steam purity.
METHODS OF STEAM PURITY MEASUREMENT
Several methods of measuring steam purity have been available and used for many years. Each offers
its own distinct advantages.
Specific conductance is one of the most commonly used methods. The specific conductance of a
sample, measured in microsiemens (µS) or micromhos (µmho), is proportional to the concentration of
ions in the sample. When boiler water is carried over in steam, the dissolved solids content of the boiler
water contaminates the steam, and steam sample conductivity increases. Measurement of this increase
provides a rapid and reasonably accurate method for determining steam purity.
One of the disadvantages of using specific conductance is that some gases common to steam (such as
carbon dioxide and ammonia) ionize in water solution. Even at extremely low concentrations, they
interfere with measurement of dissolved solids by increasing conductivity. This interference can be
appreciable in a high-purity steam sample.
For example, in a sample containing less than 1 ppm dissolved solids, specific conductance may be in
the range of 1.0-2.0 µS. The presence of any ammonia or carbon dioxide in this sample significantly
increases the conductance reading:
ammonia by 8.0-9.0 µS per ppm of ammonia
carbon dioxide by 5.0 µS per ppm of carbon dioxide
Neither of these gases is a dissolved solid. In order to obtain a proper measure of dissolved solids, the
influence of each gas must be determined, and conductivity readings must be corrected for their
presence. When the ammonia and carbon dioxide contents of the sample are known, accurate
conductivity correction curves may be obtained to allow proper corrections to be made.
Equipment is available to degas a sample prior to measurement of conductance. Hydrogen-form cation
exchange resin columns are used to reduce ammonia and amines to negligible levels. Cation
conductivity analyzers apply this technology to detect acid-producing anions, such as chlorides, sulfates,
and acetates. They also take advantage of the high conductance of solutions containing hydrogen ions.
These solutions have a conductivity several times greater than that of a solution with an equal
concentration of ions formed by a neutral salt.
In a Larson-Lane analyzer a condensed steam sample is passed through a hydrogen-form cation
exchange resin column. This resin column removes ammonia, amines, and sodium hydroxide from the
sample. The sample then flows through a reboiler, which removes carbon dioxide. Conductivity is
measured after this process and may also be measured at the analyzer inlet and ion exchange column
outlet. When conductivity is measured at all three points, a fairly complete picture of steam composition
Sodium Tracer Technique
Another commonly used method for measuring steam purity is the sodium tracer technique. This
technique is based on the fact that the ratio of total solids to sodium in the steam is the same as the ratio
of total solids to sodium in the boiler water for all but the highest-pressure (greater than 2400 psig) boiler
systems. Therefore, when the boiler water total solids to sodium ratio is known, the total solids in the
steam can be accurately assessed by measurement of sodium content. Because sodium constitutes
approximately one-third of the total solids in most boiler waters and can be accurately measured at
extremely low concentrations, this method of steam purity testing has been very useful in a large number
Sodium Ion Analyzer. The instrument most frequently used for sodium measurement is the sodium ion
analyzer. A selective ion electrode similar to a pH electrode is used to measure the sodium content of
the steam sample.
In typical operation, a regulated amount of an agent such as ammonium hydroxide is added to a
regulated amount of condensed steam sample to raise pH and eliminate the possibility of hydrogen ion
interference. A reservoir stores the conditioned sample and feeds it at a constant flow rate to the tip of
the sodium ion electrode and then to a reference electrode. The measured electrode signal is compared
to the reference electrode potential and translated into sodium ion concentration, which is displayed on a
meter and supplied to a recording device.
Good results have been reported with various sodium ion analyzers. According to the manufacturers, the
instruments operate in a concentration range of 0.1 ppb to 100,000 ppm of sodium ion with a sensitivity
of 0.1 ppb. Recalibrated on a weekly basis, these instruments are valuable for continuous, long-term
testing and monitoring.
The acceptance of the sodium ion analyzer as an accurate, reliable steam purity evaluation instrument is
evidenced by its widespread use for continuous monitoring as well as for field testing. Many steam
generating plants have switched from previously accepted methods to sodium ion analysis in order to
Although sodium ion analyzers measure total contamination, they do not show rapid changes in sodium
concentration. This is due to a lag in electrode response and the dilution effect of the reservoir, which
dampens sharp, momentary changes in conditions. Because of this, peaks that exceed boiler guarantee
limits or a known carryover range may not show up on the analyzer. This would affect interpretation of
Flame emission spectroscopy and flame spectrophotometer testing. Flame spectrophotometer
testing is much more sensitive to quick changes in operating conditions and detects peaks in solids
concentration. Flame emission spectroscopy also provides accurate measurement in the low parts per
billion range despite quick variations. Neither method is suitable for continuous, unattended monitoring.
Interpretation of Sodium Test Results. The exact ratio of sodium to dissolved solids in the boiler water
and consequently in the steam can be determined for each plant but is approximately 1:3 for most plants
(i.e., for each 0.1 ppm of sodium in the steam there is approximately 0.3 ppm of dissolved solids
Initially, the use of the sodium tracer technique for steam purity evaluation required collecting sample
bottles and transporting them to the laboratory for analysis. This technique is still a valuable tool for
steam purity measurement. Samples are gathered in special laboratory-prepared, polyethylene bottles,
and care is taken to protect against contamination.
In the preferred sampling procedure, three or four samples are drawn within a 15-minute period to
ensure representative sampling. If there are excess solids in the steam, the bottle samples are used to
define the range of the problem before implementation of an in-plant study with continuous analyzers.
Bottle samples can also be used to monitor steam purity on a periodic basis.
Experience has shown that solids levels as low as 0.003 ppm can be measured with either shipped
bottle samples or in-plant testing.
Occasionally, it is of interest to determine the amount of anions (chlorides, sulfates, acetates, etc.) in the
steam. Degassed cation conductivity provides a measure of the total anion concentration in the sample.
In addition, chloride-specific ion electrodes and ion chromatography are used to detect low levels of
REPRESENTATIVE STEAM SAMPLING
In order to ensure accurate analysis, samples must be truly representative of the steam being generated.
When the sampling procedures are not followed properly, the steam purity evaluation is of little or no
Sampling nozzles recommended by the ASTM and ASME have been in use for many years. The nozzles
have ports spaced in such a way that they sample equal cross sectional areas of the steam line.
Instructions for these nozzles can be found in ASTM Standard D 1066, "Standard Method of Sampling
Steam" and ASME PTC 19.11. Field steam studies have shown that sampling nozzles of designs other
than these often fail to provide a reliable steam sample.
Isokinetic flow is established when steam velocity entering the sampling nozzle is equal to the velocity of
the steam in the header. This condition helps to ensure representative sampling for more reliable test
results. The isokinetic sampling rate for many nozzles that do not conform to ASME or ASTM
specifications cannot be determined.
Accurate sampling of superheated steam presents problems not encountered in saturated steam
sampling. The solubility of sodium salts in steam decreases as steam temperature decreases. If a
superheated steam sample is gradually cooled as it flows through the sample line, solids deposit on
sample line surfaces. To eliminate this problem, the steam can be desuperheated at the sampling point.